Dental Use, Dental, Dental Machine manufacturer / supplier in China, offering Denture Flexible Material Injection Mould Machine /Dental Lab Equipment, 1/4′′ Cmds USB Intra-Oral Dental Intraoral Camera of Unit, 10L Stainless Steel Digital Tabletop Ultrasonic Cleaner of Dental Instrument and so on.
|Purchase Qty. / Reference FOB Price|
|1-9 Pieces||US $400|
|10-19 Pieces||US $380|
|20+ Pieces||US $360|
|Payment Terms:||T/T, Western Union, Paypal|
|Applicable Departments:||Orthodontic Department|
|Medical Device Regulatory Type:||Type 1|
The Denture Injection Unit is composed of a hot press (frame, cylinder, heater, bracket and flask etc),
and an automatic temperature control system (temperature control, timer and electric system)
AC220v/110v 50HZ ±10%
Machine × 1
Manual × 1
Fuse × 2
Denture Flask × 1
Cartridge (8g ) × 5
Cartridge (19g ) × 5
Silicon Oil × 1
Brush × 1
Hexagon Wrench × 1
Copper Gasket × 2
Charge-in Tube × 1
Discharging × 1
Structure and working principle:
1.This injection system is composed of a heater and a hot press
2.The function of the heater is that it heats the cartridge at the temperature of the 282°-288° for about
7m30s-11m, making the flexible denture material melt in the cartridge.
3.The function of the hot press is that it injects the molten material into flask which is prepared in advance.
4.The heater will make a beep sound when it finishes heating.
Installations and adjustments:
1.Fix the hot press to the worktable with 4bolts. The height of the table should be convenient for operations.
2.Put the heating furnace next to hot press. The distance between the two units should be close enough for convenient operations.
3.Before adjusting, adjust the temperature to 282° and the heating time to 7minutes and 30s.(These are factory default)
4.Connect the plug to a power supply and turn on the power. After that turn on the timer. The device will work normally if it the alarm is activated in the pre-time and the fluctuation of the temperature does not exceed 10° after during the working process.